O CMP3 é um piranômetro ISO-second-class que monitora radiação solar em todo o espectro solar. Conecta-se diretamente a um datalogger Campbell Scientific e é usado para muitas aplicações meteorologicas.Leia mais
The "-L" on a product model indicates that the cable length is specified at the time of order.
The CMP3 measures solar radiation with a high-quality blackened thermopile protected by a dome. The blackened thermopile provides a flat spectral response for the full solar spectrum range, which allows the CMP3 to be used under plant canopies or lamps, when the sky is cloudy, and for reflected radiation measurements.
The CMP3 includes a white snap-on sun shield that reduces the sensor's temperature. It also has a bubble level and adjusting leveling screws, which enable the sensor to be leveled without using a leveling base.
The CMP3 produces a millivolt signal that is measured directly by a Campbell Scientific datalogger.
Two CMP3 pyranometers can be mounted back-to-back to make a low cost albedometer. Contact Campbell Scientific for more information.
|Light Spectrum Waveband||300 to 2800 nm|
|Maximum Irradiance||2000 W/m2|
|Sensitivity Range||5 to 20 µV/W/m2|
|Operating Temperature Range||-40° to +80°C|
|Temperature Dependence||±5% (-10° to +40°C)|
|Non-Linearity||< ±2.5% (0 to 1000 W/m2)|
|Tilt Response||< ±2% at 1000 W/m2 (±80°)|
|ISO Classification||Second Class|
|Dome Diameter||3.2 cm (1.3 in.)|
|Width||7.9 cm (3.1 in.)|
|Height||6.7 cm (2.6 in.)|
|Weight||600 g (1.2 lb) with 10 m (32.8 ft) cable|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
The CMP3-L includes a bubble level and adjusting leveling screws, which allow the sensor to be leveled without using a leveling base. The CM225 Solar Sensor Mounting Stand is typically used to attach the sensor to a mast, crossarm, or pole (1.0 in. to 2.1 in. outer diameter). The CM225 consists of a rectangular plate, mounting bracket, U-bolts, washers, lock washers, and nuts. The CMP3-L should be mounted away from all obstructions or reflective surfaces that might adversely effect the measurement.
Alternatively, the CM245 Adjustable Angle Mounting Stand may be used. It includes slots that can be adjusted to any angle from horizontal to vertical.
Number of FAQs related to CMP3-L: 9
Expandir todosRecolher todos
If the dome becomes damaged, contact Campbell Scientific or Kipp & Zonen to discuss replacing the dome. The sensor will need to be sent to Kipp & Zonen for any repairs.
This may be caused by conditions in which the dome of the sensor is warmer than the thermopile. Most users see a small negative value at night. This offset, as it is commonly termed, can be important for monitoring the performance of the sensor.
The CMP3-L has a passive thermopile that produces a small analog voltage that can be equated to a representative amount of incident solar radiation. Campbell Scientific recommends connecting the leads of the CMP3-L to a Campbell Scientific data logger through a differential analog input terminal.
No. The sensor needs to be connected to a data acquisition system, such as a Campbell Scientific data logger.
Most Campbell Scientific sensors are available as an –L, which indicates a user-specified cable length. If a sensor is listed as an –LX model (where “X” is some other character), that sensor’s cable has a user-specified length, but it terminates with a specific connector for a unique system:
If a sensor does not have an –L or other –LX designation after the main model number, the sensor has a set cable length. The cable length is listed at the end of the Description field in the product’s Ordering information. For example, the 034B-ET model has a description of “Met One Wind Set for ET Station, 67 inch Cable.” Products with a set cable length terminate, as a default, with pigtails.
If a cable terminates with a special connector for a unique system, the end of the model number designates which system. For example, the 034B-ET model designates the sensor as a 034B for an ET107 system.
Not every sensor has different cable termination options. The options available for a particular sensor can be checked by looking in two places in the Ordering information area of the sensor product page:
If a sensor is offered in an –ET, –ETM, –LC, –LQ, or –QD version, that option’s availability is reflected in the sensor model number. For example, the 034B is offered as the 034B-ET, 034B-ETM, 034B-LC, 034B-LQ, and 034B-QD.
All of the other cable termination options, if available, are listed on the Ordering information area of the sensor product page under “Cable Termination Options.” For example, the 034B-L Wind Set is offered with the –CWS, –PT, and –PW options, as shown in the Ordering information area of the 034B-L product page.
Note: As newer products are added to our inventory, typically, we will list multiple cable termination options under a single sensor model rather than creating multiple model numbers. For example, the HC2S3-L has a –C cable termination option for connecting it to a CS110 instead of offering an HC2S3-LC model.
Many Campbell Scientific sensors are available with different cable termination options. These options include the following:
Note: The availability of cable termination options varies by sensor. For example, sensors may have none, two, or several options to choose from. If a desired option is not listed for a specific sensor, contact Campbell Scientific for assistance.
The LP02-L is a sensor manufactured by Hukseflux, whereas the CMP3-L is manufactured by Kipp & Zonen. Both sensors meet the ISO classification of a Second Class pyranometer.
To incorporate a sensor that is compatible with wireless sensor interfaces into a wireless network, a CWS900-series wireless sensor interface is needed, as well as an A205 CWS-to-PC interface to configure it.